Attilio Poli Managing Director at Asia Pacific Air Quality Group Pte Ltd Met One Instruments, Inc. distributor in Singapore. Monitoring the Haze at Our Own Place Sep 15, 2015 During the past few days Singapore has been affected by the haze originating from the wild fires in Sumatra (Indonesia). The concentrations of PM2.5 have often reached unhealthy levels and we all have been looking at the update of the PSI (Pollution Standard Index) data. We need to know the quality of the air that we breath The concentrations can be different from place to place and what we need to know more is how much high and unhealthy is the air that we are breathing where we are living, working, where our kids are playing. We need to know it right now It is extremely important to know that the air that we are breathing is unhealthy exactly in the moment when it happens, not later, even if it is only three hours later. Unfortunately monitoring the quality of
MSO - 6 Parameter Weather Station Remote Monitoring Share on gmail Share on email Share on blogger Share on facebook More Sharing Services 0 Office of Watershed Protection Remote Monitoring Remote Monitoring Division is responsible for generating the data that is used for billing surrounding municipalities that have entered into contractual agreements with the City of Atlanta for treatment of their wastewater. Currently, Atlanta has contractual agreements with the following jurisdictions: East Point, College Park, Hapeville, Clayton County, DeKalb County, Cobb County and Fulton County. As an example of the data produced by the fl
ANOTHER QUALITY PRODUCT FROM MET ONE INSTRUMENTS, INC. contact: firstname.lastname@example.org The BC-1050 is a two-channel, continuous, filter-based monitor that simultaneously measures black carbon “BC” and UV-equivalent particulate matter “UVPM”. Measurements are made through the principle of optical absorption. Attenuation of one light source in the near-infrared “NIR” (935 nm) and another in the ultraviolet “UV” (375 nm) are simultaneously monitored across quartz fiber filter onto which particulate matter is being deposited in real time. The attenuation of the NIR source is used to quantify BC; the attenuation of the UV source may be used to quantify UVPM (UV-equivalent particulate matter). The BC-1050 is highly sensitive: the lower limit of detection is approximately 2 ng/m3 (note 1 ng = .001 μg). The minimum time resolution of the BC-1050 is 1-minute, although 5, 10, 15, 30 and 60-minute time resolution is also available. Each BC-1050 includes: 1 BC-1050 Two Channel (935 nm and 375 nm)
Ing. Carlo Giglioni from Con.Tec Engineering with the BC 1050 Black Carbon Monitor Thank you to Ing. Giglioni for displaying Met One Instruments products at the current EAC Exhibition in Milan, Italy. Showing the BC 1050 Black Carbon Monitor, ES-642 Particle sensor and the Handheld Aerocet 831 indoor PM monitor.